How to use the temperature probe for hot air rework system (no. C1541)
The Temperature Probe for Hot Air Rework System (No. C1541) is equipped with 2 types of sensors. Before measuring temperatures as follows, connect the sensor to the thermometer ( HAKKO FG-100B, HAKKO FG-100(Discontinued) and HAKKO FG-101).
When measuring the temperature of hot air blown from the nozzle:
Correcting thermal control can be made by confirming the temperature of the hot air that is actually blown.
|Applicable products:||SMD rework HAKKO FR-801, HAKKO FR-802, HAKKO FR-803B, HAKKO 850B, HAKKO 852, HAKKO FR-830, HAKKO FM-2029|
|Sensor B:||When measuring the temperature of applicable parts (workpieces):
Rework can be performed without causing damage to the parts being reworked (QFP, etc.)
Let's look at how to measure the temperature using Sensor A! First, treat the sensor A as follows:
- Bend the end as shown in the photo
- in a contactless manner
- with a 1-2mm gap
- apply the end to the measuring point while being careful not to apply hot air to the contact section.
Be careful that hot air is only applied to the end section. Failure to do so may not allow correct temperature to be measured. * Photo of the FG-100 discontined.
Next, let's look at measurement samples using Hot Air Rework System and Preheater.
Examples of use in combination with HAKKO FG-100B and HAKKO FR-803B|
* It is also possible to use in combination with HAKKO FG-101.
|Condition where the probe end is positioned in a contactless manner with a 1-2mm gap from the nozzle end.|
B) Method for measuring hot air with Hot Air Rework System
|Hot air lineup:|
HAKKO FR-810B、HAKKO FR-811
HAKKO FG-100B、HAKKO FG-101、HAKKO FG-102
* Photo of the FG-100 discontined.
Bottom heater lineup:
HAKKO FR- 830、HAKKO FR-870B、HAKKO FR-872
C) Method for measuring hot air from Preheater
Condition where the probe end is positioned in a contactless manner with a 1mm gap from the nozzle end